Electricity Tutorial 10 - The
Cathode Ray Oscilloscope
The cathode ray oscilloscope is the electronics engineer's best friend. It is an instrument that displays alternating wave-forms. These can be simple, such as the sine-wave in alternating current. Or the waveforms can be complex, such as the pattern made by someone's voice. Cathode ray oscilloscope is a bit of a mouthful, so it is shortened to CRO.
You are not expected to describe how it operates, but you are expected to be able to set it up and interpret displays on the screen.
CRO is connected in exactly the same way as a voltmeter,
i.e. in parallel with a component.
The input resistance is very high indeed and the electron beam acts as a
pointer of negligible inertia. It
is also robust and not easily damaged by overloading.
The CRO can be used as a DC voltmeter. We get a horizontal line or a dot, depending whether the time base is on. If it is used as an AC voltmeter, it will show the sinusoidal waveform. It will, of course show more complex waveforms of audio signals, but we won't worry about these.
CRO can also be used to measure the voltage across a resistance of known value.
Therefore it can be used as an ammeter.
The CRO is shown in the picture below:
most important controls that we use are:
vertical sensitivity or y-gain setting,
calibrated in volts per centimetre (V/cm).
time base, in seconds per centimetre (s/cm).
measure the voltage on the vertical
axis. We can adjust the sensitivity by turning the knob
marked y-gain or
horizontal direction is determined by the time
base setting. We can change this by using the time
can determine the peak voltage of the AC waveform shown below.
can also read the period, which in turn allows us to work out its frequency.
peak to peak voltage is 12.8 V.
Often engineers read the peak to peak voltage off the CRO as the
determination of the 0 level is not always easy. The peak
voltage is half of the peak to peak voltage.
root mean square voltage, which we use in electrical calculations, is the
peak voltage divided by Ö(2)
the Vrms = 6.4 ¸
= 4.5 V
Look at the CRO display:
time base is set at 2 ms/cm and the y gain at 0.5 V/cm
(a) What is the peak to peak voltage?
(b) What is the peak voltage?
(c) What is the rms voltage?
(d) What is the period?
(e) What is the frequency?